- Open Access
- Open Peer Review
Medical-grade footwear: the impact of fit and comfort
© The Author(s). 2017
- Received: 27 September 2016
- Accepted: 12 December 2016
- Published: 5 January 2017
Pressure-related skin lesions on the digits are a significant cause of discomfort. Most foot pain related to ill-fitting shoes occurs in the forefoot and digital areas. Pain has been associated with poor shoe fit, reduced toe box volume, as well as contour and shape of the shoe Off-the-shelf medical-grade footwear is designed as an intervention for chronic lesions on the digits. These shoes are designed with a flexible neoprene fabric upper that is thought to reduce pressure on the forefoot and reduce discomfort associated with ill-fitting shoes. The aim of this study was to investigate the effect of an off-the-shelf, medical-grade shoe on dorsal digital pressure and perceived comfort when compared to participant’s own preferred shoe.
Thirty participants (18 females, 12 males) scored their perceived comfort whilst wearing each footwear style using a visual analog comfort scale. Dorsal digital and interdigital pressures were measured in using the WalkinSense® in-shoe pressure system. Sensors were placed on predetermined anatomical landmarks on the digits. Participants were randomly assigned the test shoe and their own shoe. Once wearing the shoe, the participants walked across a 6 m walkway and pressure data from each sensor was collected and processed to obtain peak pressure, time to peak pressure and contact time.
Participants scored the test shoe with higher comfort points than their own footwear. Overall peak pressure, pressure time integral and contact time decreased, whilst the time taken to reach peak pressure increased across all anatomical landmarks whilst wearing the test shoe. Statistically significant changes were observed for all of the measured variables relating to pressure on the medial border of the first metatarsophalangeal joint.
The test shoe provided greater comfort and reduced the amount of pressure on the forefoot. The medical-grade footwear therefore, is a viable alternative to custom made prescription footwear and is more suitable than a regular everyday shoe when treating digital lesions associated with pressure.
- Prescription footwear
- Pressure measurement
- Footwear fit
The forefoot has been highlighted as the most frequent area of pain related to footwear . Forefoot pain is commonly associated with wearing ill-fitting footwear [2, 3], causing pressure over bony prominences on the dorsum of the lesser toes, the medial aspect of the first metatarsal head or the lateral aspect of the fifth metatarsal . In the long term, it is thought that the toes can adapt to footwear restrictions by extension of the metatarsophalangeal joints and flexion of the proximal interphalangeal joints . Additionally, it has been shown that the forefoot is stiffer in habitually shod individuals and this loss of mobility may lead to greater incidences of forefoot pathology and deformity . Digital deformities are subject to hyperkeratotic lesions, clavi, or ulcerations specifically on the interphalangeal joints which are subjected to frequent friction from adjacent toes . A high proportion of corns are located on the dorsum of the 2nd, 3rd and 4th toes which often assume an elevated position as the first and fifth toes are forced to adapt to the confined area of the toe box of the shoe .
Recent studies conclude that wearing a shoe with a reduced toe box volume and shape may have poor foot health outcomes. It is suggested that this is caused by constriction of the toes which are associated with foot deformities including the development of joint pathologies and forefoot lesions [1, 9]. It is thought that shoes which do not have the capacity to accommodate the forefoot will alter the dynamics of the transverse metatarsal arch, restricting the metatarsal splay of the forefoot . Poorly fitting footwear is thought to compress the digits and alter function, eventually leading to structural changes . This compression can subsequently increase the pressure from the upper of the shoe on the toes and tissue breakdown/ulceration may occur [2, 11]. The design of the toe box related to depth and shape can impact the intensity of pressure, with a round styling on the medial border causing the least pressure and a pointed gradient to the lateral border improving pressure on the fifth digit . However, this toe box shaping is not often seen in high street shoes contributing to incorrect fit of footwear.
Ill-fitting footwear is the primary cause of foot ulceration in patients who have systemic disease, such as diabetes, with 20% of presenting ulcers being due to shoes rubbing . It has been shown that diabetic foot ulcerations are largely preventable when using custom off-loading footwear with ulcer recurrence rates found to decrease by 53% over a year follow up . Re-ulceration rates were found to be 26% among therapeutic footwear group and 83% among those who wore their own footwear in ischemic and neuropathic ulcers, from a cohort of 386 ulcerations presented in clinic . Off-loading pressure is therefore indispensable for stopping the potential progression of pre-ulcerative conditions toward lesions .
Although feet with significant deformities or delineated as high risk are recommended by NICE to be referred for bespoke footwear , there is a high dissatisfaction with prescription footwear to the point of not wearing them and they invariably become just another pair of “shoes in the cupboard” . The lack of use of prescription footwear has been associated with the size, weight, design, comfort [17–19], lack of choice and styles of the shoes prescribed . Although improved styling and newer materials used within the off-the-shelf medical-grade footwear (M-GF) has helped to increase compliance in wearing the shoe , there is still a lack of empirical data to support the use of these shoes for pressure reduction on the dorsum of the foot.
Therefore, this aims to explore the use of off-the-shelf medical-grade footwear as a pressure relieving intervention. Dorsal digital and interdigital forefoot pressure when wearing an off-the-shelf M-GF and the participants own footwear will be compared. Additionally, the comfort perception of both shoes will be assessed to evaluate if there is a difference between off the shelf M-GF and participants’ own footwear.
Thirty participants (18 females, 12 males) from a convenience sample of routine podiatry patients with an average age of 71.4 years (M = 75.7, F = 68.5), height of 1.64 m (M = 1.7, F = 1.61), weight of 78.9 kg (M = 83.22, F = 74.34) and shoe size of 7 (M = 8.5, F = 6) were recruited from a UK private podiatry clinic. Ethical approval was sought and granted from Staffordshire University Ethics Committee and informed consent was provided by each participant. Participants were included in the study if they were male or female above 50 years of age, presenting with foot pain. Participants with musculoskeletal foot deformities including hallux valgus and lesser toe deformities were included in this research. Participants with a history of current ulceration, cognitive impairment, neurodegenerative disorder, peripheral neuropathy, impaired balance, amputation, wearing the intervention footwear or had use of foot orthoses three months previously, were excluded.
The participants’ own footwear was the choice of shoe that was worn to attend the appointment, with no prior knowledge and instruction as to the suitable footwear criteria to prevent specific selection of shoe. Participants attended in a range of footwear including leather boots with fastenings and low heel profiles (maximum 2 cm), slip on leather court shoes with a heel height range of 3–5 cm, loafers and lace ups both with low heel profiles (maximum 2 cm). The participants own footwear was all purchased from a variety of high street outlets and had no defined medical features.
To ensure correct shoe size fitting, a footwear sizing measurement for each participant was taken using a Brannock device (The Brannock Device Company NY, USA) and MG-F was fitted accordingly. The participants own shoe was taken as the shoe worn to clinic and was not assessed for fit. The order of footwear testing was randomised by a card selection prior to data collection commencing.
A familiarisation period was allocated where participants were asked to walk along a 6 m walkway at a self-selected speed in the intervention shoe and the participants own shoe in the selected randomised order. Once completed for each shoe, participants were asked to rate initial comfort response by completing a 150 mm visual analogue comfort scale covering nine themes of footwear comfort . The nine areas explored included; overall comfort, heel cushioning, forefoot cushioning, side to side support, arch height, heel fit, heel height, forefoot width and shoe length. Specific words that most clearly delineate extremes were anchored at the ends of the scale with the left labelled “not comfortable all” (0 comfort point) and the right end labelled, “most comfortable imaginable” (15 comfort points).
In-shoe pressure system
Medial border of the 1st metatarsophalangeal joint
Medial border of the first interphalangeal joint
Interdigital (1/2, 2/3, 3/4, 4/5)
Proximal interphalangeal joint
5th proximal interphalangeal joint
Lateral border of the 5th metatarsal head
The comfort scale was measured by categorising nine footwear features including the overall comfort. Each characteristic was scored out of 150 mm and a total comfort score was calculated for both the M-GF and own footwear out of 1350 mm. The scores were averaged and statistically tested to find out which footwear provided the greatest level of comfort using Kruskal-Wallis test (SPSS v24 IBM USA) (p < 0.05).
In-shoe pressure system
Data was captured for a whole gait cycle with pressure analysed only during stance from the third and sixth footstep as these were identified as representing normal walking . Pressure data was processed and averaged to obtain peak pressure, time to peak pressure and contact time. Each data set was assessed for normalcy and those test conditions meeting all parametric assumptions were statistically analysed using a paired samples t-test. Wilcoxon signed rank test was completed for data sets that did not meet parametric assumptions (SPSS v24 IBM USA) (p < 0.05).
The comfort scale rating for all footwear characteristics was consistently higher for the M-GF whilst the footwear length was considered the most comfortable characteristic. There were significant differences in all of the four pressure variables whilst wearing the M-GF. There was an overall decrease in peak pressure, pressure time integral and contact time, whilst the time taken to reach peak pressure increased with the M-GF. The medial border of the first metatarsophalangeal joint (sensor 1) consistently registered statistical significant difference (p < 0.05) across all four pressure variables.
Comfort perception of the nine themes in the M-GF and own shoe. Significant difference indicate by an asterisk (*)
Sig 2-tailed (p-value)
Overall shoe comfort
Side to side support
Ball of the foot width
Ball of the foot cushioning
Mean peak pressure (PP)
Mean time to peak pressure (TtPP)
Mean contact time (CT)
Footwear choice is often made on comfort and activity . Changing footwear habits can be difficult and perceptions of ill-fitting footwear contributing to pressure related toe problems are still not fully accepted. The results from this study indicate that the use of off-the-shelf medial grade footwear (M-GF) can significantly improve comfort and in addition reduce dorsal toe pressures in a clinical population.
M-GF were ranked most comfortable in all of the defined nine footwear characteristics and were in line with previous work which attest that the design, construction and properties of footwear are important factors in footwear comfort [26, 27]. The high comfort point for the ball of foot and heel cushioning are attributed to the M-GF’s lightweight cushioning polyeurethane sole, the removable flat 4 mm foam spacer and cushioning gel insole 7 mm thick under the forefoot, not always seen in routine footwear purchased from the high street. These structural features could be important to define clinical advice for comfortable footwear when ill-fitting footwear is chosen.
Footwear length was the most significant comfortable feature scored for participants from this podiatry practice, for each shoe condition indicating that the length of the shoe was well matched to foot size. It is thought that to obtain a good fit from a shoe that a distance of 1 cm is required at the end of the toe to allow for elongation during the gait cycle . This has particular implications to lesser toe deformities which are associated with wearing shoes shorter than the foot . Additionally wearing incorrect shoe length has been associated with foot, back pain and general biomechanical imbalance . Although the participants own shoes were not measured, fitting of the M-GF shoe did include a foot sizing match. The differences in footwear comfort could have therefore been related to an improved fit from the M-GF shoe.
Footwear comfort perception is largely subjective , tactile, visual, auditory and olfactory sensations are involved in comfortable shoes selection [29, 30]. Despite initial concerns that the appearance and style of the intervention footwear may have resulted in negative perception of comfort and ultimately in low comfort scoring, participants appeared to exercise a high degree of objectivity in scoring, such that these factors did not adversely affects their scoring. This finding concurs with Williams et al.  that simply improving the appearance of shoes will improve patient compliance. Indeed it is possible that compliance may be improved by convincing patients to view therapeutic footwear as a prescription in a similar manner as the pharmacologic agents prescribed for medical needs.
Changing the perceptions of patients into believing that altering footwear style could improve pressure related foot problems is ongoing. It is clear from this study that the M-GF significantly altered all 3 pressure variables compared to the participants own footwear. However, it is not clear whether the participants’ own footwear was initially creating a pressure related foot problem that needed reducing. It is still unknown as to specific conditions for callus and other hyperkeratotic lesions to form. Yet, footwear shape has been identified as limiting foot function with the fifth toes often forced to adapt to the confined area of the toe box of footwear thus subjecting them to friction . Similarly, toe box volume can compromise toe position and a wider forefoot has been correlated with ill-fitting footwear and pain .
The shape of the toe box is also important when considering pressure reduction . The variance in shaping of the toe box allowed for a rounder wider toe box to be observed in the M-GF shoe over the participants own shoe. This altered shaping could explain why the difference in pressure, between the M-GF and participants own shoe, was reduced more at the sensors located on the medial and lateral borders of the foot rather than the sensors placed in between toes. There is some suggestion that the medial and lateral borders of the foot are the most frequent sites of foot pathology . This may however be due to the unique mechanics of the joint mechanisms of the 1st ray and the lateral column . Yet, if dysfunction in foot mechanics exists in an individual the addition of ill-fitting footwear could exacerbate pressure related problems.
The limitations of this study include the potential for bias by not blinding the researcher or participant to footwear condition as this may have influenced subjective scoring. Similarly, comfort ratings of the shoes were recorded on the day of testing after only a brief familiarisation period therefore the results presented may not be considered an accurate indicator of the degree of comfort over longer periods of wear. The distribution of data between participants showed great variability leading to large standard deviations and an uneven spread across variables. Increasing the sample size could have rectified this error and should be considered for future work. Furthermore, this study could be a useful premise for the development of a larger scale structured clinical trial to explore the material properties and the construction of the shoe in relation to pathology. Future research would benefit from analysing the effects of the footwear on participant’s specific foot pathologies, anatomical variations, gait patterns, velocity and body weight.
Introducing M-GF in place of own footwear as an intervention for digital pressure related lesions will reduce the digital pressure and improve comfort. This type of footwear should be considered as part of a relevant treatment plan when discussing footwear choice with individuals.
There is minimal research on the efficacy of appropriate non-bespoke pressure relieving footwear which clinicians can introduce to patients to purchase with confidence. Footwear catalogues such as Dr Comfort® are routinely given as part of a footwear advice with no structured research to support the use of the shoes. This study provides evidence to the efficacy of one style from the Dr Comfort® M-GF range which can be included in footwear advice protocols.
Availability of data and materials
Please contact author for data requests.
BH conducted this study as part of her MSc Clinical Biomechanics at Staffordshire University, she developed the research design and conducted all data collection and made the initial draft of the manuscript. HB participated in the design of the study, statistical analysis and wrote the final versions of the manuscript. AG analysed and processed the raw data. NC contributed to the final versions of the manuscript. All authors read and approved the final manuscript.
The authors declare there are no competing interests. The shoes used in this study were independently sourced, no funding was received from Dr Comfort®.
Consent for publication
A consent form was used as part of the studied and all participants gave informed consent for anonymised data to be used for publication.
Ethics approval and consent to participate
Ethics approval for this project was granted by Staffordshire University ethics committee. A full peer reviewed ethics application was submitted and approved.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Paiva De Castro A, Rebelatto JR, Aurichio TR. The relationship between foot pain, anthropometric variables and footwear among older people. Appl Ergon. 2010;41(1):93–7.View ArticlePubMedGoogle Scholar
- Macfarlane DJ, Jeffcoate WJ. Factors contributing to the presentation of diabetic foot ulcers. Diabet Med. 1997;14(10):867–70.View ArticlePubMedGoogle Scholar
- Veves A, Murray HJ, Young MJ, Boulton AJM. The risk of foot ulceration in diabetic patients with high foot pressure: a prospective study. Diabetologia. 1992;35:660–3.View ArticlePubMedGoogle Scholar
- Lavery LA, Armstrong DG, Vela SA, Quebedeaux TL, Fleischli JG. Practical criteria for screening patients at high risk for diabetic foot ulceration. Arch Intern Med. 1998;158(2):157–62.View ArticlePubMedGoogle Scholar
- Coughlin MJ. Mallet toes, hammer toes, claw toes and corns: causes and treatment of lesser-toe deformities. Postgrad Med. 1984;75:191–8.View ArticlePubMedGoogle Scholar
- Kadambande S, Khurana A, Debnath U, Bansal M, Hariharan K. Comparative anthropometric analysis of shod and unshod feet. Foot (Edinb). 2006;16(I4):188–91.View ArticleGoogle Scholar
- Thomas JL, Blitch EL, Chaney DM, Dinucci KA, Eikmeier K, Rubin LG, Stapp MD, Vanor JV. Diagnosis and treatment of forefoot disorders. Section 1: digital deformities. J Foot Ankle Surg. 2009;48(2):230–8.View ArticlePubMedGoogle Scholar
- Menz HB, Morris ME. Footwear characteristics and foot problems in older people. Gerontology. 2005;51:346–51.View ArticlePubMedGoogle Scholar
- Branthwaite H, Chockalingam N, Greenhalgh A. The effect of shoe toe box shape and volume on forefoot, interdigital and plantar pressures in healthy females. J Foot Ank Res. 2013;6:28.View ArticleGoogle Scholar
- Louwerens JW, Schrier JCM. Lesser Toe Deformities. G Bentley (ed) European Surgical orthopaedics and Traumatology. The EFORT Text 2014; 3649–3501.Google Scholar
- Apelqvist J, Larsson J, Agardh CD. The influence of external precipitating factors and peripheral neuropathy on the development and outcome of diabetic foot ulcers. J Diabet Comp. 1990;4(1):21–5.Google Scholar
- Macfarlane RM, Jeffcoate WJ. Factors contributing to the presentation of diabetic foot ulcers. Diabet Med. 1997;14:867–70.View ArticlePubMedGoogle Scholar
- Uccioli L, Faglia E, Monticone G, Favales F, Durola L, Aldeghi A, Quarantiello A, Calia P, Menzienger G. Manufactured shoes in the prevention of diabetic foot ulcers. Diabetes Care. 1995;18(10):1376–8.View ArticlePubMedGoogle Scholar
- Edmonds ME, Blundell MP, Morris ME, Maelor TE, Cotton LT, Watkins PJ. Improved survival of the diabetic foot: the role of a specialized foot clinic. Q J Med. 1986;60:763–71.PubMedGoogle Scholar
- NICE CG 10 Type 2 diabetes foot problems: Prevention and management of foot problems (2004) http://www.nice.org.uk/guidance/cg10. Accessed on 07/09/2016
- Willliams AE, Meacher K. Shoes in the cupboard: the fate of prescribed footwear? Prosthet Orthot Int. 2001;25:53–9.View ArticleGoogle Scholar
- de Boer IG, Peeters AJ, Ronday HK. Assistive devices: usage in patients with rheumatoid arthritis. Clin Rheumatol. 2009;28(2):119–28.View ArticlePubMedGoogle Scholar
- Williams AE, Nester CJ, Ravey MI. Rheumatoid arthritis patients’ experiences of wearing therapeutic footwear - a qualitative investigation. BMC Musculoskelet Disord. 2007;1(8):104.View ArticleGoogle Scholar
- Williams AE, Nester CJ. Patient perceptions of stock footwear design features. Prosthet Orthot Int. 2006;30(1):61–71.View ArticlePubMedGoogle Scholar
- Bowker P, Rocca E, Arnell P, Powell E. A study of the organisation of orthotic services in England and Wales. UK: Report to the Department of Health; 1992.Google Scholar
- Menz HB, Auhl M, Frescos N, Munteanu SE. Effectiveness of off-the-shelf, extra-depth footwear in reducing foot pain in older people: a randomized controlled trial. J Gerontol A Biol Sci Med Sci. 2015;70(4):511–7.View ArticlePubMedGoogle Scholar
- Nancarrow S. Footwear suitability scale: a measure of shoe fir for people with diabetes. Australas J Podiatr Med. 1999;33:57–62.Google Scholar
- Mundermann A, Nigg BN, Stefanyshyn DJ, Humble RN. Development of a reliable method to assess footwear comfort during running. Gait Posture. 2002;16:38–45.View ArticlePubMedGoogle Scholar
- Healy A, Burgess-Walker P, Naemi R, Chockalingam N. Repeatability of WalkinSense® in shoe pressure measurement system: a preliminary study. Foot (Edinb). 2012;22(1):35–9.View ArticleGoogle Scholar
- Branthwaite HR, Chockalingam N, Pandyan A, Khatri G. Evaluation of lower limb electromyographic activity when using unstable shoes for the first time: a pilot quasi control trial. Prosthet Orthot Int. 2012;10:1177.Google Scholar
- Hennig EM, Milani TL. Pressure distribution measurements for evaluation of running shoe properties. Sportverletz Sportschaden. 2000;14(3):90–7.View ArticlePubMedGoogle Scholar
- Nigg BM, Nurse MA, Stefanyshyn DJ. Shoe inserts and orthotics for sports and physical activities. Med Sci Sports Exerc. 1999;31(7 suppl):421–8.View ArticleGoogle Scholar
- Kong PW, Bagdon M. Shoe preference based on subjective comfort for walking and running. J Am Podiatr Med Assoc. 2010;100(6):456–62.View ArticlePubMedGoogle Scholar
- Goonetilleke RS, Luximon A. Designing for comfort: a footwear application, Computer-aided ergonomics and safety conference. 2001.Google Scholar
- Au EYL, Goonetilleke RS. A qualitative study on the comfort and fit of ladies’ dress shoes. Appl Ergon. 2007;38(6):687–96.View ArticlePubMedGoogle Scholar
- Branthwaite H, Chockalingam N, Greenhalgh A, Chatzistergos P. The impact of different footwear characteristics, of a ballet flat pump, on centre of pressure progression and perceived comfort. Foot (Edin). 2014;24(3):116–22.View ArticleGoogle Scholar
- Zipfel B, Berger LR. Shod versus unshod: the emergence of forefoot pathology in modern humans? Foot (Edin). 2007;17(4):205–13.View ArticleGoogle Scholar
- Lundgren P, Nester C, Liu A, Arndt A, Jones R, Stacoff A, Wolf P, Lundberg A. Invasive in vivo measurement of rear-mid-and forefoot motion during walking. Gait Posture. 2008;28(1):93–100.View ArticlePubMedGoogle Scholar