Volume 7 Supplement 1
The use of a robotic gait simulator for the development of an alignment tool for lower limb prostheses
© De Raeve et al; licensee BioMed Central Ltd. 2014
Published: 8 April 2014
An innovative tool to optimise the configuration and alignment of lower leg prostheses based on individual comfort needs of the patient will be developed in this project.
This tool meets the demand of prosthetists to optimise and objectify the dynamic alignment of transtibial prostheses. Nowadays, the prosthetists mainly rely on their own expertise and experience.
Current methods to align the prosthesis start with a static alignment of transtibial prosthesis. This method does not take the individual patient comfort into account. Afterwards, adjustments to the alignment are done by trial and error. This is a time-consuming and exhausting activity for both prosthetist and patient.
Subsequently, the 3D movement of the shank was recalculated to Euler angles to be used as kinematic input for an industrial robot with 6 degrees of freedom (KUKA) (Figure 1B). An energy consumption system with spring enables us to simulate kinematic and kinetics of prosthetic gait. To objectively measure the pressure, an artificial stump was developed to connect the lower limb prosthesis to the robot (Figure 1C). This artificial stump, which consists of SLS-3D printed bones and soft tissue (silicon) allows us to objectively measure the pressure between stump and socket.
For different alignments, forces (AMTI), motion trajectories and contact pressure between stump and socket (mFLEX) were registered simultaneously during the unroll of the prosthetic foot. These measurements are translated into comfort parameters during post-processing.
Results and discussion
Using our robotic gait simulator, the effect of prostheses alignment on patient comfort is mapped. The biggest advantage of this approach is the possibility to test numerous and extreme alignments, without the need for test subjects. After validation with gait analyses, this data is translated in an algorithm which will be converted into a useful method as a tool for the prosthetist.
The outcome of this project will lead to an improved efficiency and effectiveness of the alignment process, and contribute to the quality of life of the amputee.
Clinical Trial Center - CTC 55509
This study is funded by TETRA IWT-Vlaanderen (project Ticoon-IWT120144).
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.