- Poster presentation
- Open Access
The effect of a subject-specific AFO on the muscle activation during gait of a test subject suffering from a hemiparetic anterior muscle insufficiency in the lower leg
© Creylman et al; licensee BioMed Central Ltd. 2012
- Published: 10 April 2012
- Test Subject
- Selective Laser Sinter
- Sinter Technique
- Musculoskeletal Model
- Patient Anatomy
An Ankle Foot Orthosis (AFO) is commonly used in clinical practice to assist gait of patients with different pathologies. The flexibility of the AFO depends on different design characteristics while specific mechanical requirements of the AFO are correlated with patient anatomy and pathology. To this day, the correlation between AFO-design and patient pathology is mainly based on the orthopaedic technician’s experience.
The aim of this study is to investigate the influence of the stiffness of an AFO on the muscle-activation pattern of a subject suffering from an anterior muscle insufficiency of the lower leg using a personalized musculoskeletal model.
Test subject was a 40-year old male suffering from a hemiparetic anterior muscle insufficiency of the lower leg. A musculoskeletal model of the lower limbs with 23 degrees of freedom and 92 muscles was scaled in OpenSim to match the test subject’s anthropometric data . Muscle-definitions were adapted to simulate the patients’ pathology.
Marker trajectories of an AFO-gait were used to calculate kinematic parameters and muscle-activation during gait using the musculoskeletal model. The AFO was simulated as an angle-dependent torque around the ankle with a neutral angle of 0°. The stiffness of the AFO was varied between 150 and 350 Nm/rad in steps of 25 Nm/rad.
Results showed alternating muscle-activation in the affected lower leg with increasing AFO-stiffness. No clear correlation could be found between AFO-stiffness and muscle-activation.
An optimal AFO-stiffness in terms of muscle-activation can be selected for the patient. It must be kept in mind that these results are based on a calculation with a neutral AFO ankle position of 0°, and changing this neutral angle, affects the results.
- Delp SL, Anderson FC, Arnold AS, Loan P, Habib A, John CT, Guendelman E, Thelen DG: OpenSim: Open-Source Software to Create and Analyze Dynamic Simulations of Movement. IEEE Trans Biom Eng. 2007, 54: 1940-1950.View ArticleGoogle Scholar
- Pallari JHP, Dalgarno KW, Munguia J, Muraru L, Peeraer L, Telfer S, Woodburn J: Design and Additive Fabrication of Foot and Ankle-Foot Orthoses. Proceedings of the 21st Annual International Solid Freeform Fabrication Symposium – An Additive Manufacturing Conference, 9-11. 2010, August , Austin, Texas, USAGoogle Scholar
- Muraru L, Creylman V, Pallari J, Willemsen R, Vander Sloten J, Peeraer L: Evaluation of functional parameters of ankle foot orthoses for different materials and design characteristics. Proceedings of the 13th World Congress of the International Society for Prosthetics and Orthotics, May 10-15. 2010Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.