- Oral presentation
- Open Access
Comparison of gait data using two different protocols for ankle joint kinematics
© Krauss et al; licensee BioMed Central Ltd. 2008
- Published: 26 September 2008
- Joint Angle
- Gait Analysis
- Plantar Flexion
- Altman Plot
- Normal Weight Subject
Gait analysis is an important instrument in various fields of clinical research and its protocols are intended to make kinematics interpretable for clinicians.
Although they use the same nomenclature for joint angles, different protocols produce different results . The purpose of this study was to compare gait events of the ankle joint to determine differences between two protocols.
Two different protocols were used to quantify distinctive kinematic variables in the stance-phase of barefoot walking at a normal speed:
(1) A functional approach (FA), assuming a ball-and-socket joint at the ankle .
(2) A prediction approach (PA) based on anatomical studies , differentiating between upper and lower ankle joint.
A single comprehensive marker-set was defined allowing the use of exactly the same gait cycles for both protocols. 10 healthy normal weight subjects (mean = 27 y, s = 5.8) were analyzed on two consecutive days M1 and M2. Landmark definition was done by the same physiotherapist for all subjects. A 12-camera Vicon MX40 system system collected data at 100 Hz. Five force plates were used to detect gait events. Peak eversion, frontal plane range of motion (ROM) and peak plantar flexion of the ankle joint were extracted from the curve data. At least 5 trials were averaged for each subject and measurement day, respectively.
Agreement between the two methods was quantified using the Bland & Altman Plot .
Based on the preliminary data, there does not seem to be a relation between the magnitude of the mean values of the two protocols and the magnitude of their differences. Furthermore, differences between protocols are similar for both days.
The use of different gait analysis protocols for the description of ankle joint kinematics yields different results for both absolute joint angles and ROM. This should be considered when results from different studies are compared. The Bland & Altman Plot illustrates differences between methods in a very comprehensive way, incorporating diverse information in one single plot. Final results will incorporate limits of agreement to allow an estimate of the range in which 95% of the differences can be expected.
- Ferrari , et al: Gait Posture. 2008.Google Scholar
- List , et al: J Biomech. 2006, 39 (S1): S550-10.1016/S0021-9290(06)85266-8.View ArticleGoogle Scholar
- Inman : Bul Pros Res. 1969, Spring: 130-145.Google Scholar
- Bland , Altman : Stat Meth Med Res. 1999, 2 (8): 135-60. 10.1191/096228099673819272.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd.