- Oral presentation
- Open Access
The role of shear stress in the aetiology of diabetic neuropathic foot ulcers
© Giacomozzi et al; licensee BioMed Central Ltd. 2008
- Published: 26 September 2008
- Shear Force
- Peripheral Neuropathy
- Motion Capture
- Plantar Pressure
- Motion Capture System
Several biomechanical studies have been conducted in the last decade to investigate the aetiology of plantar ulcers in presence of Diabetes (D) and Peripheral Neuropathy (PN) . Great renewed interest has been lately showed towards shear stresses during gait [2, 3]. In this case, the major difficulty lays in technology, since it is still hard to obtain reliable and meaningful measurements by using the available measurement systems. The authors' validated methodology  is here re-proposed, and the results of its further applications to the analysis of PN shear stress are here briefly reported.
A compound instrument was made at ISS (Rome, Italy) by superimposing a resistive pressure platform (4 sensors/cm2; pressure resolution 15.2 kPa; 100 Hz) on a Bertec force plate (force resolution 2 N, moment resolution: 0.3 Nm), and a mathematical model was implemented whose input are local vertical forces and global shear forces. Output of the model are estimated local shear forces and free moment . The measurement system was used to investigate 61 barefoot D patients (34 PN, 21 matched controls = C). Three subareas – heel, metatarsals and hallux – were geometrically identified by making reference to the lines at 40% and 70% of the total length perpendicular to the bisecting line of the foot .
A similar prototype was then developed at the University of Padova (Italy), and the mathematical procedure replicated. In this case a 6-cameras BTS motion capture system (60–120 Hz) synchronized with two Bertec force plates and integrated with two Imago resistive pressure platforms (1.56 sensors/cm2; 150 Hz) were used. The foot subareas were defined by projecting the anatomical landmark positions onto the plantar pressure footprint . 38 patients were analyzed: 10 C,14 diabetics (D), 14 PN. Three subareas were again selected, rearfoot, midfoot, forefoot .
The above studies confirm the need of a deep biomechanics analysis of the diabetic foot including vertical and shear forces during gait. The results from the two studies were in a good agreement, absolute differences mainly due to partially different subarea selection and adopted technologic solution. Further reliable measurement instrumentation and foot models should be validated and applied, also including in-shoe measurements, and the analysis of motor tasks more demanding than level walking.
- Cavanagh P, et al: Edited by: Bowker JH, Pfeifer MA. 2001, The Diabetic Foot, St. Louis, 125-196.Google Scholar
- Ledoux WR: Foot and Ankle Motion Analysis. Edited by: Harris GF,s et al. 2008, CRC Press, 317-345.Google Scholar
- Giacomozzi C, et al: Diabetes Care. 2002, 25: 1451-1457. 10.2337/diacare.25.8.1451.View ArticlePubMedGoogle Scholar
- Giacomozzi C, Macellari V: IEEE Trans On Rehab Eng. 1997, 5 (4): 322-330. 10.1109/86.650285.View ArticleGoogle Scholar
- Sawacha , et al: Gait & Posture. 2006, 24: S2-S3. 10.1016/j.gaitpost.2006.09.018.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd.